How to Use the And ua-701 Digital Blood Pressure Meter: A Step-by-Step Guide
If you are looking for a reliable and easy-to-use device to monitor your blood pressure at home, you might want to consider the And ua-701 digital blood pressure meter. This device is clinically validated and has a large display and memory function. In this article, we will show you how to use the And ua-701 digital blood pressure meter manual and give you some tips on how to get accurate readings.
And ua-701 digital blood pressure meter manual
What You Need to Know Before Using the And ua-701 Digital Blood Pressure Meter
Before you start using the And ua-701 digital blood pressure meter, there are some things you need to know:
The device measures your blood pressure and pulse rate using the oscillometric method, which detects the pressure changes in your artery as your heart beats.
The device comes with a cuff that fits arm circumferences from 22 to 32 cm. You can also purchase a large cuff that fits arm circumferences from 31 to 45 cm.
The device can store up to 30 measurements for one user. You can also delete the stored measurements if you want to clear the memory.
The device has an irregular heartbeat indicator that flashes if it detects an irregular rhythm during the measurement.
The device has a WHO blood pressure classification indicator that shows you how your blood pressure compares to the World Health Organization standards.
The device has a battery indicator that shows you when the batteries are low. The device requires four AA batteries to operate.
How to Use the And ua-701 Digital Blood Pressure Meter Manual
To use the And ua-701 digital blood pressure meter manual, follow these steps:
Insert the batteries into the battery compartment on the back of the device. Make sure to align the polarity correctly.
Connect the cuff tube to the air jack on the left side of the device.
Wrap the cuff around your upper arm, about 2-3 cm above your elbow. Make sure the cuff is not too tight or too loose, and that the air tube is not twisted or bent.
Sit comfortably on a chair with your feet flat on the floor and your arm resting on a table at heart level. Relax for a few minutes before taking a measurement.
Press the START/STOP button to turn on the device. The device will automatically inflate the cuff and start measuring your blood pressure and pulse rate.
Wait until the device displays your results on the screen. The device will also show you your WHO blood pressure classification and irregular heartbeat indicator if applicable.
Press the START/STOP button again to turn off the device. The device will automatically deflate the cuff and store your measurement in its memory.
Repeat steps 3-7 for another measurement if desired. Wait at least one minute between measurements.
How to Get Accurate Readings with the And ua-701 Digital Blood Pressure Meter
To get accurate readings with the And ua-701 digital blood pressure meter, follow these tips:
Avoid eating, drinking, smoking, exercising, or bathing for at least 30 minutes before taking a measurement.
Avoid taking a measurement when you are stressed, anxious, or in pain.
Avoid talking or moving during a measurement.
Take your measurements at the same time of day, preferably in the morning and evening.
Compare your readings with those taken by your doctor or nurse, as they may differ due to different measurement methods or conditions.
How to Troubleshoot the And ua-701 Digital Blood Pressure Meter
If you encounter any problems with the And ua-701 digital blood pressure meter, refer to this troubleshooting guide:
No display or weak displayBatteries are low or exhaustedReplace with new batteries
Error message E1Cuff is not wrapped correctlyWrap the cuff properly around your arm
Error message E2 0efd9a6b88